Venturi tube

A Venturi tube (or simply a venturi) is a section of piping consisting of an inlet converging conical section leading to a small diameter cylindrical section called the throat, followed by a diverging conical section leading to a cylindrical exit (see the adjacent drawing).

The volumetric flow rate, Q (e.g., m3/s), of a fluid flowing through the venturi at any given point is the product of the cross-sectional area, A (e.g., m2) at that point multiplied by the fluid's linear velocity, v (e.g., m/s), at that point. Since the volumetric flow rate is the same at any point within the venturi, the linear velocity of the fluid traveling through the smaller diameter throat of the venturi must increase which results in a decrease of pressure due to the conservation of energy, one of the conservation laws.

The gain in kinetic energy resulting from the increased linear velocity in the throat is offset (i.e., balanced) by the decrease of pressure in the throat. The reduction in pressure which occurs when the fluid flows through the throat is called the Venturi effect and is named after the Italian physicist Giovanni Battista Venturi (1746 - 1822) who first observed the effect.

Thus, referring to the adjacent drawing, the pressure p2 and the cross-sectional area A2 in the throat are smaller than the pressure p1 and the cross-sectional area A1 in the cylindrical inlet section. The linear velocity v2 in the throat is higher than the linear velocity v1 in the inlet section.

A venturi tube may also consist of non-cylindrical ducting rather than piping.

Incompressible fluid
Venturi tubes are used to accurately measure the volumetric flow of an ideal fluid, meaning that the fluid satisfies these conditions:


 * The fluid is nonviscous: There is no internal friction between layers of the fluid.
 * The fluid is incompressible: The density of the fluid is constant (as is the case with liquids)
 * The fluid flows without any turbulence: No element of the fluid has any angular velocity about its center, so there are no eddy currents present.
 * The fluid flow is steady: The velocity, density and pressure at each point in the fluid does not change with time.

To obtain the volumetric flow rate of an incompressible fluid through a venturi, we start with this common form of the Bernoulli equation for fluid flow:


 * $$p_1 + \tfrac {1}{2}\, \rho_1\, v^2_1 + \rho_1\, g\, h_1 = p_2 + \tfrac {1}{2}\, \rho_2\, v^2_2 + \rho_2\, g\, h_2$$

where ρv2/2 denotes kinetic energy, ρgh denotes gravitational potential energy and:


 * {|border="0" cellpadding="2"


 * align=right|$$p$$
 * align=left|= the fluid's static pressure
 * align=right|$$\rho$$
 * align=left|= the fluid's density
 * align=right|$$v$$
 * align=left|= the fluid's linear velocity
 * align=right|$$g$$
 * align=left|= the gravitational acceleration
 * align=right|$$h$$
 * align=left|= the height of a given point (in the fluid flow) above any reference plane designated as having zero height
 * }
 * align=left|= the gravitational acceleration
 * align=right|$$h$$
 * align=left|= the height of a given point (in the fluid flow) above any reference plane designated as having zero height
 * }
 * }

For an incompressible fluid with a constant density and a venturi with a horizontal centerline, the equation simplifies to:


 * $$p_1 + \tfrac {1}{2}\, \rho\, v^2_1 = p_2 + \tfrac {1}{2}\, \rho\, v^2_2$$

or to:


 * $$p_1 - p_2 = \tfrac {1}{2}\, \rho\, (v_2^2 - v_1^2)$$

Since the volumetric flow rate Q is the same at all points in the venturi by the continuity equation:


 * $$Q = A_1\, v_1 =A_2\, v_2$$

then:
 * $$p_1 - p_2 = \tfrac {1}{2}\, \rho \left(\frac {Q^2}{A^2_2} - \frac {Q^2}{A^2_1}\right)$$

which can be re-arranged to:


 * $$Q = A_2 \sqrt {\frac {2\, (p_1 - p_2)}{\rho \left (1 - \left [\frac {A_2}{A_1} \right ]^2 \right)}}$$

The actual flow rate through a venturi tube is seldom equal to the theoretical flow rate of an ideal fluid and usually turns out to be less than the theoretical flow rate. In order to account for this difference, a discharge coefficient C is introduced to obtain the final volumetric flow rate equation for an incompressible fluid flowing through a venturi tube with a horizontal centerline:


 * $$Q = C A_2 \sqrt {\frac {2\, (p_1 - p_2)}{\rho \left (1 - \left [\frac {A_2}{A_1} \right ]^2 \right)}}$$

where:


 * {|border="0" cellpadding="2"


 * align=right|$$Q$$
 * align=left|= the volumetric flow rate of the fluid
 * align=right|$$C$$
 * align=left|= the discharge coefficient = (actual flow rate)&thinsp;/&thinsp;(theoretical flow rate)
 * align=right|$$A$$
 * align=left|= the cross-sectional area of the venturi in the indicated section of the venturi
 * align=right|$$p$$
 * align=left|= the fluid's static pressure in the indicated section of the venturi
 * align=right|$$\rho$$
 * align=left|= the fluid's density
 * }
 * align=left|= the fluid's static pressure in the indicated section of the venturi
 * align=right|$$\rho$$
 * align=left|= the fluid's density
 * }
 * }

One method for determining the discharge coefficient is to calibrate the venturi tube on a test stand which is expensive and time consuming. A considerable number of test calibrations have been performed on various venturi tube geometries. As a result, the ASME Standard MFC-3M-1989 provides these discharge coefficients for their specified venturi tube geometry (per the diagram at the top of this article) and for three types of fabrication:


 * For venturi tubes with a welded convergent cone: C = 0.984 (valid for fluid flow Reynolds numbers between 2 × 105 and 6 × 106
 * For venturi tubes with an as-cast convergent cone: C = 0.984 (valid for fluid flow Reynolds numbers between 2 × 105 and 6 × 106
 * For venturi tubes with a machined convergent cone: C = 0.995 (valid for fluid flow Reynolds numbers between 2 × 105 and 2 × 106

Compressible fluid
For compressible fluids (i.e., a gases), the density is not constant. It will vary from the density ρ1 at the venturi inlet to the density ρ2 at the venturi throat. The equation for the mass flow rate of a compressible fluid through a venturi may be expressed as:


 * $$\dot {m} = Y C A_2\, \sqrt \frac {2\, \rho_1\, (p_1 - p_2)}{1 - \beta^4}= Y C A_2\, \sqrt \frac {2\, \rho_1\, (p_1 - p_2)}{1 - \left ( \frac {d}{D} \right )^4 }$$

The expansion factor Y allows for the change in gas density as the gas expands adiabatically from the venturi inlet pressure to the reduced pressure at the venturi throat. It may be calculated by using this equation:


 * $$Y = \sqrt {\, r^{2/k} \left(\frac{k}{k-1}\right) \left(\frac {1 - r^{(k-1)/k)}} {1-r} \right ) \left (\frac{1 - \beta^4}{1 - \beta^4 r^{2/k}} \right)}$$

where:


 * {|border="0" cellpadding="2"


 * align=right|$$\dot m$$
 * align=left|= mass flow rate
 * align=right|$$Y$$
 * align=left|= gas expansion factor
 * align=right|$$C$$
 * align=left|= the discharge coefficient
 * align=right|$$A_2$$
 * align=left|= cross-sectional area of the venturi's throat section
 * align=right|$$\rho_1$$
 * align=left|= the gas density in the venturi's inlet section
 * align=right|$$p_1$$
 * align=left|= the gas static pressure in the venturi's inlet section
 * align=right|$$p_2$$
 * align=left|= the gas static pressure in the venturi's throat section
 * align=right|$$\beta$$
 * align=left|= d&thinsp;/&thinsp;D, the ratio of the venturi's throat diameter d to the venturi's inlet diameter D
 * align=right|$$r$$
 * align=left|= p2&thinsp;/&thinsp;p1
 * align=right|$$k$$
 * align=left|= the specific heat ratio of the gas
 * }
 * align=right|$$\beta$$
 * align=left|= d&thinsp;/&thinsp;D, the ratio of the venturi's throat diameter d to the venturi's inlet diameter D
 * align=right|$$r$$
 * align=left|= p2&thinsp;/&thinsp;p1
 * align=right|$$k$$
 * align=left|= the specific heat ratio of the gas
 * }
 * align=right|$$k$$
 * align=left|= the specific heat ratio of the gas
 * }

Other uses for a venturi
Venturi tubes are used to create a low pressure or a partial vacuum in a great many applications (other than measuring flow rates) including the examples below as well as many others:


 * Injectors or ejectors using steam, water, air or other motive fluid (liquid or gas) in a venturi to create a low pressure zone that draws in and entrains a suction fluid (liquid or gas).
 * Water aspirators using water as a motive fluid to produce a partial vacuum.
 * Atomizers using air as a motive fluid to spray paint, perfume or other liquids.
 * Eductors using high-pressure water to create a partial vacuum to draw in and mix firefighting foam concentrate with the water.
 * Carburetors using a gasoline combustion engine's intake air stream as the motive fluid to entrain and mix gasoline into intake air.
 * Compressed air as the motive fluid to create the vacuum for industrial vacuum cleaners.
 * Venturi scrubbers used in flue gas desulfurization systems.
 * Sand blasters using compressed air to draw in fine sand in and mix it with air.
 * A scuba diving regulator to assist the flow of breathing air once it starts flowing.
 * In Venturi masks used in medical oxygen therapy.
 * The De Laval nozzles used in rocket engines are essentially venturi tubes.